The AccuTOF DART couples the simple, rapid sampling of the DART ion source with the high-resolution, accurate mass capability of the AccuTOF time-of-flight mass spectrometer. The example shown here is taken from one of our product presentation videos. A business card was spotted with trace TNT (nominal m/z 227) and then sampled intact using the AccuTOF DART. Without high-resolution, accurate mass capability, it would be easy to misinterpret the tallest peak as TNT, when it is actually due to the ubiquitous environmental contaminant myristic acid. The high-resolution, accurate mass spectra collected with AccuTOF DART allow you to distinguish individual sample components in your crude sample mixtures, as well as unambiguously identify them.
View Product Presentation Video of AccuTOF DART Analyses
Back to AccuTOF DART Product Page